8. CHECK AND ADJUSTING
Please perform check and adjusting regularly. First check, and then make adjustments accordingly.
Horizontal Rotation Grade Error
• How to check
1Set up tripod approximately 50 meters
away from a wall, and set the instrument on
level with the X1 facing the wall.
2While pressing the key, turn ON the
power. (Only the RL-SV1S is operable).
[CaLIb] will flash on the X axis screen. *1)
3Press the key. (Hereafter, the RL-SV1S
and RC-50 become operable.)
8.1 Check and Adjust Horizontal Rotation
Staff or Wall
X- laser beam
X- laser position