8. CHECK AND ADJUSTING
The screen will be in the X-axis check and adjusting mode.
Auto leveling on the RL-SV1S is complete and the laser will emit.
4Turn the LS-80L to the high precision mode.
5Detect the center of the laser beam on the wall with the LS-80L and mark it. (X1)
6Press the key.
7Loosen the centering screw and rotate the RL-SV1S 180°, and tighten the screw to secure.
The RL-SV1S X+ surface will face the wall. When rotating the RL-SV1S, ensure that the
instrument height is not misaligned.
The RL-SV1S auto leveling is complete and the laser will emit.