8. CHECK AND ADJUSTING
• How to adjust
1According to step 9 of the horizontal rotation check, press the keys* to move the
laser beam between X1 and X2.
* When adjusting the Y axis direction, press the keys.
2Press the key.
* Display during arrow key
operation (p. 38)
Adjustment for X axis is complete.
Turn OFF the power.
Exceeding the range of adjustment.
12. ERROR DISPLAY(p. 66)
The RL-SV1S is calculating the correction value. Do not touch the RL-SV1S until
[End] is displayed. (If you touch it, you will need to readjust.)