8. CHECK AND ADJUSTING
7Measure the difference X between the standard line and laser beam.
8If X is within 1 mm, no adjustment is required. If the difference exceeds 1 mm, move on to
the next adjustment.
Vertical calibration and adjustment
1While pressing the key, press the key.
Press the key.
2Move the RL-SV1S in the direction of the arrow to align the standard line and laser beam.
3Press either the key or the key to align the laser beam with the LS-80L indicator
position. (check with the buzzer sound from the LS-80L)