About The Meter 2
2.4 FRONT OF THE METER(Continued)
In the Run Mode, press the MENU button to terminate the current measuring process
and enter you into the Configuration Mode.
Only if you have not installed the lockout jumpers on the main board.
In the Configuration Mode, press the MENU button to store changes in the nonvolatile
memory and then advance you to the next menu item.
If you hard reset (press the MENU button followed by the RESET button) or power off/on
the meter, it shows RST , followed by PROC .
In the Run Mode, press the RESET button to reset tare, if any. The meter shows T.RST
and returns to the Run Mode.
In the Configuration Mode, press the RESET button once to review the previous menu.
Press the RESET button twice to perform a hard reset and return to the Run Mode.
In the Peak Mode, press the RESET button to reset peak values. The meter shows
PK.RS and returns to the Run Mode.
In the Setpoint Mode, press the RESET button to reset the latched setpoint. The meter
shows SP.RS and enters the Run Mode.
When in setpoint or Configuration Mode, if the meter shows 9999 or -1999
with all flashing digits, the value has overflowed. Press the 䊱/MAX button
to start a new value.