You will be asked to Save the Changes. Press “Y” for Y)es.
Then, press “A” for A)ccept.
The program will then automatically return to the normal mode, displaying data that is
5.4.5 CALIBRATING THE ANALOG OUTPUTS
If you have a precise milliammeter, you can easily calibrate the three 4 to 20 mA analog
outputs. In order to utilize this convenient capability, connect the meter in series with
one of the analog outputs and the load.
Press the “Escape” key on your keyboard, which will interrupt the data stream and give
you the choice of D)isplay or C)al Outputs. Select “C”.
In order to calibrate Output 1, press “1”. You will then be able to insert the desired
current value at the ¼ FS (25% of Full Scale) point and at the 3/4FS (75% of Full Scale)
point. For 4 to 20 mA, 1/4FS is 8 mA, and 3/4FS is 16mA.
Insert these values, pressing “Enter”
after each one. You will then see a
range of what the milliammeter should
read in order to attain the desired
Then select “X” for eX)it. The system
will now return to the normal data