5 CALIBRATION AND CALIBRATION KIT
To perform the optional directivity calibration, connect a
LOAD standard to the test port as shown in figure 5.7 and
perform measurement using Load (Optional) softkey.
The instrument status bar will indicate Calibration in
progress... when the measurement is in progress. On
completion of the measurement a check mark will appear
in the left part of the softkey.
To complete the calibration procedure, click Apply.
This will activate the process of calibration coefficient
table calculation and saving it into the memory. The error
correction function will also be automatically enabled.
To clear the measurement results of the standards, click
This softkey does not cancel the current calibration. To
disable the current calibration turn off the error correction
function (See section 5.2.10).
Note You can check the calibration status in channel status bar
(See table 5.5) or in trace status field (See table 5.6).