5 CALIBRATION AND CALIBRATION KIT
To display detailed information on characterization, use
Characterization Info softkey.
To perform full two-port calibration, use 2-Port AutoCal
To perform full one-port calibration, use 1-Port AutoCal
Then select the port number.
5.8.3 User Characterization Procedure
User characterization of ACM is required in case of ACM connectors modification by
the use of adapters. The characterization is performed for the new ACM configuration,
which includes adapters. To ensure calibration accuracy it is not recommended to
disconnect and reconnect the adapters back.
Before you perform the user characterization of the ACM, perform full two-port
calibration of the Analyzer in configuration of the test ports compatible with the
configuration of ACM ports.
Connect the ACM to the Analyzer test ports, and connect the USB port of the ACM to
the USB port of the computer.
Select user characterization 1 to 3 using Characterization
Select manual or automatic orientation of the ACM
using Orientation softkey.
It is recommended to select AUTO orientation.
Perform characterization using Characterize ACM
After the ACM measurement is completed, the following dialog box will appear: