3 GETTING STARTED
In the current example Agilent 85032E calibration kit is used.
To select the calibration kit, use the following softkeys:
Calibration > Cal Kit
Then select the kit you are using, from the table in the
bottom of the screen.
To perform full 1-port calibration, execute measurements of the three standards. After
that the table of calibration coefficients will be calculated and saved into the memory of
the Analyzer. Before you start calibration, disconnect the DUT from the Analyzer.
To perform full 1-port calibration, use the following
Calibration > Calibrate > Full 1-Port Cal
Connect an OPEN standard and click Open.
Connect a SHORT standard and click Short.
Connect a LOAD standard and click Load.
To complete the calibration procedure and calculate the
table of calibration coefficients, click Apply softkey.
Then connect the DUT to the Analyzer port again.