4 MEASUREMENT CONDITIONS SETTING
4.4.4 Trace/Channel Activating
The control commands selected by the user are applied to the active channel or the
active trace, respectively.
The border line of the active channel window is highlighted in light color. The active
trace belongs to the active channel and its title is highlighted in inverse color.
Before you set a channel or trace parameters, first you need to activate the channel or
To activate a trace/channel, use the following softkeys:
Display > Active Trace/Channel
Then activate the trace by entering the number in the
Active Trace softkey or using Previous Trace or Next
Active channel can be selected in similar way.
4.4.5 Active Trace/Channel Window Maximizing
When there are several channel windows displayed, you can temporarily maximize the
active channel window to full screen size. The other channel windows will be hidden,
but this will not interrupt the measurements in those channels.
Similarly, when there are several traces displayed in a channel window, you can
temporarily maximize the active trace. The other traces will be hidden, but this will not
interrupt the measurements of those traces.