4 MEASUREMENT CONDITIONS SETTING
To enable/disable active channel maximizing function,
use the following softkeys:
Display > Active Trace/Channel > Active Channel
To enable/disable active trace maximizing function, use
the following softkeys:
Display > Active Trace/Channel > Active Trace
Note Channel and trace maximizing function can be controlled
by a double mouse click on the channel/trace.
4.5 Stimulus Setting
The stimulus parameters are set for each channel. Before you set the stimulus
parameters of a channel, make this channel active.
Note To make the measurement more accurate, perform
measurements with the same stimulus settings as were
used for the calibration.
4.5.1 Sweep Type Setting
To set the sweep type, use the following softkeys:
Stimulus > Sweep Type
Then select the sweep type:
■Lin Freq: Linear frequency sweep
■Log Freq: Logarithmic frequency sweep
■Segment: Segment frequency sweep
■Power Sweep: Power sweep